How to judge tube furnace using thermocouple _ instrument
First determine the appearance of the thermocouple wire is no problem, it is good or bad, have to go through testing to determine. Will stay thermocouple wire wear measurement thermocouple special porcelain sleeve, with standard platinum-rhodium thermocouple in the tube furnace, the hot side insert tube furnace in a porous soaking with metal in the cylinder made of nickel. Put their compensation the cold end of the wire in the ice water to keep zero degrees Celsius in the container. Hold the tube furnace at the high allowable temperature of the thermocouple, and keep the stable range. By this time with the tested qualified potentiometer, measure standard thermocouple and under test thermocouple thermoelectric potential difference and record. According to the records of thermoelectric potential difference, check grade table find out their corresponding temperature, if the thermocouple out-of-tolerance under test, can be judged to be unqualified.
During the exhibition season, the company team, with company letter product thermocouple, thermal resistance electronic temperature and other products, visited our company's European and South American customers, meet with old customers at the exhibition, new products are welcomed by old customers; should be invited, visit customer company, deepen The understanding of both sides is for further cooperation.