The ozone power _ instrument are described
The ozone power supply part is mainly produce the uv lamp need to high voltage power supply, at the same time produce circuit boards need to + 15 v dc. Uv lamp light control part mainly is to control the uv light lamp current in the range of constant allows, can automatically adjust the too high, too low, if not adjust a uv lamp to life, there is a red light on the panel lights, you should replace the uv lamp. Standard uv detection and sampling ultraviolet detection part is a key part,. Photoelectric sensor to ultraviolet light signal is converted into a voltage signal, and then the two operational amplifiers, signal amplification, gave LOG100 calculated after processing, the display output. Simulation of 0 ~ 20 ma with the size of ozone concentration is a linear relationship.
During the exhibition season, the company team, with company letter product thermocouple, thermal resistance electronic temperature and other products, visited our company's European and South American customers, meet with old customers at the exhibition, new products are welcomed by old customers; should be invited, visit customer company, deepen The understanding of both sides is for further cooperation.