Proficient in industrial commonly used thermocouple type thermocouple temperature sensing element (
The resistor body)
The structure and characteristics.
From the thermocouple temperature measurement principle, the measured the change in temperature is directly measured by thermocouple resistance change, therefore, the wires of a thermocouple body such as a variety of wire resistance change will affect temperature measurement.
As to eliminate the influence of the lead resistance with the three wire or four wire system.
Thermocouple is by thermal components (
The resistor body)
, wire, insulation materials, stainless steel casing combination solid body, its diameter is generally from 2 to 8 mm, achieve more xiaoke phi is 0.
Compared with conventional type thermocouple, it has the following advantages of (1) small volume, no internal air gap, thermal inertia, the measured value after small;
(2) good mechanical properties, resistance to vibration and shock.
(3) can bend, easy to install (4) long service life.
End thermocouple temperature sensing element resistance wire winding by special processing, close to the face in the thermometer, it is compared with general axial thermocouple, can more accurately and quickly to reflect the actual temperature of end face, is suitable for measuring temperature bearing and other parts of the face.
Flameproof thermocouple junction box flameproof thermocouple with special structure, its shell internal explosive mixed gas by affected by sparks or electric arc explosion limit within the junction box, that wouldn't make production field explosion.
Flameproof thermocouple can be used in the Bla ~ B3c zone with explosion danger place temperature measurement.
Thermocouple temperature measurement system of thermocouple temperature measurement system in general by thermocouple, connecting wires and display instrument, etc.
Must pay attention to the following two points: (1) the thermocouple and the index number of the display instrument must agree (2) in order to eliminate the influence of the connection wire resistance change, must use three wire connection.