Why can't the compensation to the wire connected to the thermocouple temperature over 100 ℃ _ instrument
We all know that temperature thermocouple temperature difference is the use of electric potential, however, due to the cold end temperature and display and control instrument, and temperature difference between the general use of thermoelectric potential and thermocouple close to the potential of cheap metal instead of hot electrode signal, this is the compensation conductor, this kind of compensation conductor should be 1, copper -
Compensation K dividing nickel chrome - constantan KC
Nickel silicon thermocouple 2, copper -
Copper and nickel SC compensation S dividing platinum rhodium - 10
Platinum thermocouple both compensation conductor should be belong to type of compensation, the main problem is to compensate the material after the thermoelectric potential of more than 100 degrees, error increases, at work, thermocouple cold junction temperature shall not be greater than 100 degrees, even using ptfe or glass fiber insulating sheath, the compensation temperature also shall not be greater than 100 degrees, if you are using PVC insulating sheath, its working environment temperature
Thermocouple cold end.
Shall not be greater than 70 degrees.
Some inexpensive metal thermocouple and thermal electrode materials consistent conductor do compensation conductor, this kind of compensation conductor should be called expanded compensation wire, they have 1, nickel chromium -
Nickel silicon KX extension compensation degree nickel chrome - K
Nickel silicon thermocouple 2, nickel chromium -
Constantan EX extended compensation degree nickel chrome - E
Constantan thermocouple 3, - of iron
J dividing iron - constantan JX extension compensation
Copper constantan thermocouple, 4 -
Copper nickel T indexing copper - TX extension compensation
Copper and nickel thermocouple such compensation conductor thermoelectric potential and thermocouple completely consistent, so, in the temperature of the thermocouple cold junction compensation is not restricted, if using ptfe jacket, you can use an ambient temperature of 250 degrees, the use of glass fiber sheath environment temperature is 500 degrees, if use PVC sheathed with environment temperature shall not be greater than 70 degrees.
In addition, B dividing thermocouple due to potential very weak at low temperatures, below 400 degrees without precision, so B dividing platinum rhodium - 30
Platinum rhodium thermocouple 6 without compensation wire, double core is commonly used in copper sheath line.