4-20mA output PT100 Temperature sensor with transmitter
100mm or customized
1. The thermal resistance should be mounted vertically on the horizontal or vertical pipe as much as possible.
A protective sleeve should be installed for easy access and replacement.
2. When measuring the temperature inside the pipe, the length of the component should be on the center line of the pipe (ie, the insertion depth of the protection pipe should be half of the pipe diameter).
3, when the temperature dynamic table is installed, the opening size should be appropriate, the installation should be beautiful and generous.
4. Use high temperature resistant cable or high temperature compensation line in high temperature zone.
5. Select different measuring components according to different temperatures. Generally, the thermal resistance is selected when the measured temperature is less than 400 °C.
6, the wiring should be reasonable and beautiful, the needle indication should be correct.
Hello, please leave your name and email here before chat online so that we won't miss your message and contact you smoothly.